Liquid peracetic acid-based disinfectant for invasive and non-invasive instruments
PerActionTM One Shot is a high-level disinfectant developed with a single 5% peracetic acid concentration and by adding stabilizers and corrosion inhibitors, to be reconstituted in tap water in a 1:36 ratio in order to obtain a solution with a peracetic acid content above 1100 ppm.
After diluting 5% peracetic acid in water, the disinfectant solution is ready for use and must be used within about 2 hours for its biocidal activity to be carried out.
The product was developed for high-level disinfection of surgical instruments and probes in automatic equipment that does not reuse a disinfectant solution for multiple cycles.
PerActionTM One Shot is available in 5L jugs.
PerActionTM One Shot is suitable to carry out any top quality instrument washing and disinfection process: surgical probes, surgical instruments, various or heat-sensitive material, automated endoscope washers, fibre optics tools.
PerActionTM One Shot is used in:
Top quality disinfection of any DM labelled device or instrument for a manual application or to be used in an instrument or probe washing machine.
Thanks to its different packaging, it complies with many different functions and requirements of clinics, surgeries and sterilisation centres.
To disinfect dental surgical tools, manually washed or in automated instrument washing machines.
Operating instructions of automated endoscope washing machines
Add the 5 litre canister content to the endoscope washing machine.
Program the machine to directly draw 28.6 ml of peracetic acid for every litre of water.
The solution’s antiseptic activity is active within 15 minutes.
The canister content should be entirely used 8 days after being opened.
Operating instructions for the manual disinfection
Dilute the PerAction™ One Shot solution with a 1:36 ratio (1 litre disinfectant in 35 litres of water).
5 litres of PerAction™ One Shot provide for a total 180 litre disinfecting solution.
The solution is ready to use: not waiting time for its activation and it is active for two hours.
After washing the instruments, completely immerse them in PerAction™ One Shot for 15 minutes.
Then remove the instruments, according to an antiseptic mode and rinse with sterile water.